If you’re having trouble with the water pump within your Tuttnauer autoclave you need to discover what the issue is right away and fix it. No water or perhaps a water degree that’s as well low means your machine isn’t functioning properly and won’t be completely sterilizing your equipment. And that means you’re essentially out of company before you get the problem resolved.
If you’ve constantly delivered your autoclave out for repair whenever there’s a difficulty then you know just how costly these fixes can be. You have work charges, costs for parts, and don’t forget about the shipping and handling fees. You can nearly buy a new machine when all is considered and done! If you’re having trouble with the water within your Tuttenauer autoclave, below are a few problem solving tips that might assist you to solve the issue your self.
When the water is flowing in the wrong direction, check both pipes connected to the pump. It may be which you just must change them.
In the event the water pump is making noise check to determine if the installation bracket is loose or damaged. You may just need to tighten it down or replace it. If it’s not the bracket the issue is probably in the water pump alone and you’ll need a new one.
When the water pump helps to keep coming a fuse, initially look for pinched or twisted wires that might be resulting in a short. If that’s not the issue, transform the water pump off and disconnect the cables from your number 1 and two terminals in the solid state communicate. Transform the water pump back on and when the fuse blows once again the problem is with either the water water pump or the pump capacitor and you’ll have to change each.
When the pump won’t switch off you will want a voltage/ohmmeter to test the communicate. Read among TP13 and TP1. A 10 to 12 voltage DC transmission is needed to tell the relay to transform away along with a to 1 signal is required to inform it to change on. If the reading through is anything else it indicates there is a downside to the control circuit.
In the event the signal is proper, replicate the method at the relay. And if you obtain an incorrect transmission there then you’ll must replace the Ajunc board.
If your pump won’t switch on, the very first thing you must do is check the fuse and change it if needed. Following, look for pinched or twisted cables leading from your solid state relay for the pump. If ejxawh doesn’t assist, transform the power in your autoclave away and disconnect the wires on terminals 1 and two in the relay. Now, connect the two wires with each other and transform the water pump back on. In the event the water pump nevertheless doesn’t work then you’ll need to change it.
Whilst your autoclave might become a ‘machine’ and it performs an important, lifestyle-conserving functionality, it’s still, after all, just a big box with a few fundamental working parts. It’s typically easy sufficient to get the issue. And most of the time you can correct it your self!